Have a question or want to order over the phone? Call us at 1-800-285-2807.
The Ultima 20 TENS – is a dual Channel device with twenty modes of operation (8 Manual and 12 PRESET). It is simple to operate, and has a protective cover over the controls. Its pulse width and pulse frequency are adjustable. Nerve Stimulator TENS Machine With Twenty Modes is durable, ergonomic, has always a potent defense against pain. Tens Neuromuscular Stimulator can be used for nerve pain relief and discomfort. This electric nerve stimulator comes ready-to-use with 12 reusable electrodes. Nerve stimulator is a device used to treat chronic pain, and pain associated with active or post-traumatic injury unresponsive to other standard pain therapies.
The electrical nerve stimulation device is applied to the surface of the skin at the site of the pain. Just remove the sleek black nerve stimulation unit from its professional case, insert the battery, apply the electrodes, and plug in the lead wires. Turn on the nerve stimulator unit which has 20 modes (8 Manual and 12 PRESET), to adjust the pulse width (how strong the current is) and the frequency (how many times the current is delivered over time) and you can enjoy a pain free day while you take it anywhere you go! TENS neuromuscular therapy stimulates sensory nerves to block pain signals,stimulate endorphin production to help normalize sympathetic function and has a biphasic, low volt current. Electrical nerve stimulator comes with 5 warranty against mechanical defects.This type of stimulation is characterized by biphasic, low volt, current and selectable parameters such as pulse rate and pulse width. TENS stimulates sensory nerves to block pain signals, stimulate endorphin production to help normalize sympathetic function. A prescription from your physician or physical therapist is required for this device.
The handheld device comes ready-to-use with 12 FREE Electrodes (each disposable electrode lasts 10-15 treatments), 5 year limited warranty against mechanical defects, Industrial Strength 9V Battery, Pair of lead wires, carrying case, belt clip, and user manual.
Nerve Stimulation Package includes:
(1) TENS Unit
(3) packs of electrodes (4/pack)
(2) Lead wires
(1) Users manual
(4) Self Adhesive Pads with Connectors (Size: 40mm x 40mm)
(2) Alkaline AA 1.5V (LR6) Batteries
Available in Black and White Colors
# 1) Labels MANUAL and PRESET on the LCD display, will switch on according the program.
# 2) Labels ACUTE and CHRONIC will switch on according to the Pulse Width (50 us = Gate Control, then most suitable toACUTE pain – Above 50 us = Endorphins then most suitable to chronic pain), regardless the PW is manually or automatically adjusted.
# 3) Labels ACUTE and CHRONIC will switch on according to the Pulse Width (50 us = Gate Control, then most suitable toACUTE pain – Above 50 us = Endorphins then most suitable to chronic pain), regardless the PW is manually or automatically adjusted.
# 4) Labels ACUTE and CHRONIC will switch on according to the Pulse Width (50 us = Gate Control, then most suitable toACUTE pain – Above 50 us = Endorphins then most suitable to chronic pain), regardless the PW is manually or automatically adjusted.
# 5) Symbols will show the occurring phase (if blinking) or the passed phase (if permanently ON) of the PRESET programs.
# 6) Labels CONT, MOD and BURST have not to be shown during the PRESET programs.
# 7) When mode changes, the intensity will automatically go down to zero.
# 8) When the unit is turned on, it will automatically enter the mode that the unit had worked in before the unit was turned off.
# 9) When the treatment timer is set, it will begin to count down one minute by one minute; once it counts down to zero, the unit will automatically shut off.
# 10) When the pulse rate, pulse width or intensity is being adjusted, the auto-repeat function is available, when the key is held down over 1 second, it will automatically increased or decreased 1 step per 1/4 second.By pressing “PR-” key and “PW-” key simultaneously, the form of the stimulation wave is changed (for manual programs only) and the change is directly felt by the user.
# 11) By pressing “MODE” key and “PR-” key simultaneously for 4 seconds, the unit will be locked or unlocked in toggle. While the unit is locked, all parameters cannot be adjusted except timer and intensity. By removing batteries or pressing ON/OFF key cannot change the status of locked/unlocked.
# 12) The treatment time will be accumulative recorded when the output level is above zero, by pressing “MODE” key and “TIME” key for 4 seconds the treatment time accumulated in minute can be displayed, or return back to the previous normal display in toggle; by pressing “MODE” key and “PW-” key for 4 seconds will clear the treatment time to zero.
# 13) When the unit intensity levels are 0 on both channels, and it has not been in use for 5 minutes, the unit will be shut off automatically.